Test Alkalinity with Confidence!
The Kamoer KH Carer is an automatic testing device measuring KH values (Alkalinity) in an aquarium. It includes manual and automatic detection functions. Users can easily view the test results and set the test parameters in the Android or iOS app. The equipment can adjust the KH value according to the test results automatically.
It includes a 2.8-inch display screen
Features high-precision pump heads
Automatically completes the test and adds KH reagent according to the results
Supports two detection modes: automatic test and manual test
Supports pump flow calibration and pH calibration
Includes a real-time clock, which runs automatically according to the set parameters, and the power down parameters are not lost
Support iOS and Android with remote WiFi and local Bluetooth control
Support the solution bottle capacity monitoring function, and push the alarm when the solution bottle capacity is insufficient
Support the function of alarm push
Anatomy of the Kamoer KH Carer
1-Saltwater sampling interface, connected to saltwater tank
2-Add KH detection reagent interface and connect to KH detection reagent barrel
3-Add KH inlet and connect KH bucket
4-Add KH outlet, connected to sea tank
5-Waste water outlet, connected to waste liquid drum
6-Communication interface, standby
7-Power supply DC12V 1.9A
10-pH electrode fixing cover
12-Magnetic stirring rotor
It includes automatic and manual working modes
1-24 tests per day in automatic mode
Reagent bottle remaining quantity monitoring function
Alarm push function
Automatic supplement KHfunction
Calibrate pump and function
1. Install the beaker
After receiving the equipment, the user should put the beaker on the equipment for use.
Gently pull out the pH electrode and put the magnetic stirring rotor into the
Put the pH electrode and waste liquid discharge pipe into the beaker, then plug them into the fixed position, and the installation is completed.
The position of the waste liquid discharge pipe depends on the inner wall of the beaker to prevent the magnetic rotor from touching the waste liquid discharge pipe when rotating.
2. Configuration solution
This step can be ignored if the KH detection reagent that can be used directly is purchased. The following describes the dilution method of KH detection reagent concentrate.
The electronic scale measures 980G RO 980G water corresponds to 980ml water. RO water can be obtained from the water dispenser. The reagent barrel should be peeled on the electronic scale before taking water.
Pour a bottle of KH concentrate into 980G water and gently shake the reagent
Pour the mixed reagent of water and concentrated solution in the second step into the concentrated solution bottle and rinse the concentrated solution bottle
Pour the rinsed liquid back into the reagent barrel
Repeat steps (3) and (4) once to complete the dilution of the concentrate.
Download the App
Apple users enter the App Store application store, Android users enter the Google Play application store, search for "Kamoer Remote", find the application download
CoralVue, Inc. and Kamoer hereby guarantee this product to be free from defects in material and workmanship for a period of:
Warranty is activated on the date of purchase.
This Standard Limited Warranty applies and is limited as follows:
To the product only as long as it remains in the possession of the original purchaser.
To the product that has not been subject to accident, misuse or abuse.
To the product that has not been modified, altered, defaced, or had repairs made or attempted by other than CoralVue, Inc.
That CoralVue be immediately notified in writing within ten (10) days of first knowledge of defect by owner or his agent.
That CoralVue, Inc. shall be given first opportunity to make any repairs, replacements or corrections to the defective construction within a reasonable period of time.
Under no circumstances shall CoralVue and/or the Manufacturer be liable by virtue of this warranty or otherwise for damage to any person or property whatsoever for any special, indirect, secondary or consequential damages of any nature however arising out of the use or inability to use because of the product defect.